3-strip loading capacity or break-apart single and double strips.
Carriers come in two styles 3x12 for break-apart microwell strips and an optional 3x8 for non-break-apart microwell strips.
Bichromatic optics. Standard model includes four filters six filters are available for a nominal charge.
Easy to use with LCD touch screen with color graphics display.
Easy loading on-board graphics printer.
On-board software calculates single & multi-point calibration curves several regression calculations and a variety of cutoff and control options.
Large non-volatile memory stores more than 50 user-entered tests.
NRTL and CE certified.
Stat Fax® readers use IAD filters
The Awareness Technology 303 Microstrip Reader is a fully automatic reading unit and uses biochromatic optics. The standard model includes 4 filters with the option of 6 filters at a nominal charge.
The onboard data reduction calculation includes regression to plot curve fitting and cut off absorbance calculation. This is made simple with step my step user prompting and the new larger screen display and graphical capabilities.
The Stat Fax 3200 Microplate Reader has a large- non-volatile memory to store over 100 user tests which can be transferred using parallel and serial outputs or printed via the optional onboard printer.
The Awareness Technology Stat Fax 303 Microstrip Reader Specifications
Alpha numeric- 16 character LCD
Thermal- dot matrix- 20 character per line- plus graphics
16-key- domed membrane switch- enunciating
115V or 230V AC- 50-60Hz (switch selectable)
Output only- custom cable available- 2400 baud- 1 start- 8 data- 1 stop- no parity- no handshake
Z80A or Z180
Battery supported RAM- holds more than 32 tests and curves
-0.20 to 3.0 Absorbance units (A)
+/- 1 percent or better
Drift of no more than 0.005A in 8 hours
Tungsten lamp with lamp saver feature
Standard Wave Lengths
405- 450- 492 and 630 nm
6-VIS: 405- 450- 492- 545- 600- and 630nm
6-UV: 340- 405- 450- 492- 545- and 630nm
Multiple cavity hard coat interference. 10nm half bandpass
Standard 96-well microtiter plates- or strip trays
L x W x D inch
9 x 12 x 3
It is never a good thing when things don't go quite to plan. If your laboratory instrument is giving you trouble you have come to the right place! In three simple steps you can be on your way receiving a quote on the repair of your broken lab instrument.
Step 1. Tell us a little about the instrument that is giving you trouble:
If you purchased your instrument from GMI in the past, It will have a Product Identification Number, this can be easily found by locating the PID sticker. The PID Sticker resembles something like this.